A very simple machine, very easy to use, but essential to vestibular rehabilitation.
|On a shelf||Remote Controler||Touchpad||On the ceiling|
This stimulator is made up of a reflective faceted ball enlightened by a light source with adjustable intensity and accuracy thus allowing us to vary the area of the patches which are reflected.
This ball can move around both axes of rotation in order to enlighten the whole machine according to the 3 planes: the entire visual display unit is thus scanned.
Each of the 2 axes is driven by a very quiet electronic motor to avoid the patient’s focusing his attention on the light source.
Those motors can work either in one way or another with adjustable speed. We can also combine the movement of both axes. Motor and light source can be operated by remote control.
The visual distortion is a power unit and ocular reaction to the shifting of the whole visual field. This reaction is a visual distortion whose slow stage is in direction of the stimulus and quick one is opposite to the shifting.
The optokinetic stimulation interferes with the VOR (Vestibular Ocular Reflex) by means of visual distortion, which in turn may confound postural control and the vestibular spinal reflex. The velocity and direction of this optokinetic nystagmus can be adjusted, changing the direction and speed while the ball is in rotation.
This optokinetic stimulation allows us to detect visual dependency (in populous environment, transportation ills…) but also to cure balance disorders (like vestibular function sleeping), and strengthen balance function. Moreover, if the Stimulopt is coupled with the Multitest Platform (BalanceQuest) the detection and cure of the disorder can be quantified.
This product is utilized by everybody working on balance disorders, like physiotherapists, ENTs, audiologists, neurologist…
Click here to get to the manuals download page.
Click here to get to the brochures download page.
• Base dimension : 18 X 12 X 12 inches (46x30x30 cm)
• Weight : 9 lbs (4 kg)
• Power : 110 or 220 VAC
• Light source : LED
The faceted ball can rotate along 2 axis, and direction and speed can be controlled separately.
• infrared remote
• PC controlled via Multitest Platform
• via a wall mounted LCD touch screen display
• Cycle 0:
Manual Mode (Ball and Z axis speed/direction).
• Cycle 1:
Same as Cycle 0, but ball direction changes every X seconds (set by user between 1 and 59 seconds).
• Cycle 2:
Same as Cycle 1, but timing of direction change is randomly defined.
• Cycle 3:
Speed is randomly defined. No direction change.
• Cycle 4:
Speed is randomly defined. Direction changes randomly